Connect Origin Software to Sensors via
USB Data Acquisition Hardware SOFTWARE > ORIGIN SOFTWARE >
OriginLabs and instruNet
The instruNet-to-Origin Widget populates an
Origin spreadsheet while digitizing in realtime, as pictured to the right. This requires instruNet software ≥ v3.6, Origin ≥ v8.6 and a 64bit Windows ≥ 7 computer.
Import Post Acquistion or Realtime?
If you want to learn how to digitize with instruNet World and save a text file to be Imported by Origin, post-acquisition, please see
Excel Import (which is similar to Origin Import). If you want to
digitize directly into Origin, while data is being acquired in realtime, please consider the instruNet-to-Origin widget described here.
In order to run the instruNet-to-Origin widget, all of the following must be installed on your computer. If you are not sure these requirements have been satisfied, try running the widget, as
described below, and if it runs ok, you are probably ok.
instruNet software ≥ 3.6,
available at www.instrunet.com/d.
This will install a 64bit native version of "windows/system32/iNet32.dll", which will talk to Origin.
Origin "instruNet_to_Origin.opx" file
and place it onto the Windows desktop.
Run Origin software and wait for it's window to open. Drag the "instruNet_to_Origin.opx" file onto the Origin window to permanently install this widget into Origin. An alert will appear that states "The package has been installed". Also, a toolbar with one button named "iNet" will appear in the Origin window. Drag this to the side of the window to dock it, as shown to the right.
If instruNet hardware is not connected to this computer and you want
to simulate an instruNet i555
hardware device (where you receive random numbers
instead of real a/d values), changed file name
"...Documents\ instruNet\ End User\ iNet32 Demo Off.txt" to
"...iNet32 Demo On.txt".
If you want to later disable simulation, rename the file back to "..Off". Simulation still requires instruNet
software to be installed on the computer, which one can do for free via
Quit application software that might be operating instruNet, run Origin, and click the "iNet" button. This will run
the widget, as pictured below.
Press Channels to
channels and to enable them for digitizing. To enable a channel for digitizing, click to make rectangle red, as shown above and described here. After setting up your channels, close the Network window by clicking it's upper-right close box.
Press Setup to set samples-per-second-per-channel and number-of-points-digitized-per-channel. Typically, one sets Number-of-Scans to 1.
PressSave to save the setup. Everything you did in the Channels and Setup areas is saved in a .prf file. Make sure you don't save to the Program Files folder, since this is a protected area.
Next time you run this software, you must press Load to load back your settings.
PressStart to start digitizing, and Stop to stop it.
For more details on how to use this widget, press Help.
Press the Channels button, select 2 channels for digitizing (i.e. click in small rectangle to right of blue Quick Setup button, it will turn red), and close the instruNet Network window via upper-right close box.
Press the Start button and watch a new spreadsheet window appear and fill with instruNet data, as
shown in the picture at the top of this webpage. When finished, you can save or delete the spreadsheet file.
Close the instruNet widget by clicking in the upper-right corner of the widget window. One can re-open by clicking the "iNet" button.
If you cannot get the widget to operate properly, quit Origin, run instrunet world software (START / instruNet / instruNet World.exe), and make sure instruNet hardware is seen by your computer.